-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20150926_160411_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.615054 to 36.142170
-each separately from 8s to 36s and from 141881us to 615343us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.943741 in 1 packets - #/No: 18/18
      16.582508 in 1 packets - #/No: 36/18
MIU packet time from 2015-09-26T16:04:22.615.054.000 to 2015-09-26T16:04:50.142.170.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.943741 in 1 packets - #/No: 18/18
      16.582508 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4542755 to 37.4541998
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.9999703 in 1 packets - #/No: 18/18
      16.9999540 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      C0 -> 35.6°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20150926_160411_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.0693293 to 10.0696183
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.9437110  -- OK for SL --
   MIU time from 2015-09-26T16:04:22.615.054.000 to 2015-09-26T16:04:22.615.343.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.943741

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.0133293 to 21.0136183
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5824630  -- OK for SL --
   MIU time from 2015-09-26T16:04:33.559.084.000 to 2015-09-26T16:04:33.559.373.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.582508

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.5960813 to 37.5963703
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-09-26T16:04:50.141.881.000 to 2015-09-26T16:04:50.142.170.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-09-26T16:04:22.614.913.750
   Max SL corrected packet time: 2015-09-26T16:04:50.142.029.750
   Min SL corrected XO rel time: 8.6150555
   Max SL corrected XO rel time: 36.1421709
   TM values from -73 to 50 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-09-26T16:04:22.614.913.750 (XO: 8.6150555, t=34.0°C)  Rank: 00006
   TM values: Max =    21  Min =   -35  Ave =    -4.71  Std =     7.73
   Min TM value:-35 at an offset of 131.113 usec -> 2015-09-26T16:04:22.615.044.862 (XO: 8.6151866)
   Max TM value:21 at an offset of 130.225 usec -> 2015-09-26T16:04:22.615.043.975 (XO: 8.6151857)
   Max dTM/dt [1/us]::2.47e+02 at an offset of 130.300 usec -> 2015-09-26T16:04:22.615.044.050 (XO: 8.6151858)
   Min dTM/dt [1/us]::-2.41e+02 at an offset of 62.213 usec -> 2015-09-26T16:04:22.614.975.962 (XO: 8.6151177)

 - Block number: 2 starting at 2015-09-26T16:04:33.558.943.750 (XO: 19.5590850, t=34.0°C)  Rank: 00006
   TM values: Max =    50  Min =   -45  Ave =    -4.73  Std =     6.28
   Min TM value:-45 at an offset of 205.038 usec -> 2015-09-26T16:04:33.559.148.788 (XO: 19.5592900)
   Max TM value:50 at an offset of 204.088 usec -> 2015-09-26T16:04:33.559.147.838 (XO: 19.5592891)
   Max dTM/dt [1/us]::4.88e+02 at an offset of 204.038 usec -> 2015-09-26T16:04:33.559.147.788 (XO: 19.5592890)
   Min dTM/dt [1/us]::-4.55e+02 at an offset of 146.288 usec -> 2015-09-26T16:04:33.559.090.037 (XO: 19.5592313)

 - Block number: 3 starting at 2015-09-26T16:04:50.141.740.750 (XO: 36.1418819, t=34.0°C)  Rank: 00005
   TM values: Max =    31  Min =   -73  Ave =    -4.73  Std =     6.93
   Min TM value:-73 at an offset of 126.025 usec -> 2015-09-26T16:04:50.141.866.775 (XO: 36.1420079)
   Max TM value:31 at an offset of 126.925 usec -> 2015-09-26T16:04:50.141.867.675 (XO: 36.1420088)
   Max dTM/dt [1/us]::6.96e+02 at an offset of 126.063 usec -> 2015-09-26T16:04:50.141.866.812 (XO: 36.1420079)
   Min dTM/dt [1/us]::-6.18e+02 at an offset of 125.988 usec -> 2015-09-26T16:04:50.141.866.737 (XO: 36.1420079)
