-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20150918_140852_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.465243 to 40.308000
-each separately from 1s to 40s and from 307711us to 740688us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.274867 in 1 packets - #/No: 18/18
      18.567023 in 1 packets - #/No: 36/18
MIU packet time from 2015-09-18T14:08:56.465.243.000 to 2015-09-18T14:09:35.308.000.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.274867 in 1 packets - #/No: 18/18
      18.567023 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.3997095 to 41.3996084
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.9999481 in 1 packets - #/No: 18/18
      18.9999508 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20150918_140852_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.8649523 to 2.8652413
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.2748150  -- OK for SL --
   MIU time from 2015-09-18T14:08:56.465.243.000 to 2015-09-18T14:08:56.465.532.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.274867

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.1400563 to 23.1403453
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.5669750  -- OK for SL --
   MIU time from 2015-09-18T14:09:16.740.399.000 to 2015-09-18T14:09:16.740.688.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.567023

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.7073203 to 41.7076093
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-09-18T14:09:35.307.711.000 to 2015-09-18T14:09:35.308.000.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-09-18T14:08:56.465.102.750
   Max SL corrected packet time: 2015-09-18T14:09:35.307.859.750
   Min SL corrected XO rel time: 1.4652440
   Max SL corrected XO rel time: 40.3080017
   TM values from -18 to 10 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-09-18T14:08:56.465.102.750 (XO: 1.4652440, t=35.6°C)  Rank: 00003
   TM values: Max =     5  Min =   -17  Ave =    -4.85  Std =     3.01
   Min TM value:-17 at an offset of 96.500 usec -> 2015-09-18T14:08:56.465.199.250 (XO: 1.4653405)
   Max TM value:5 at an offset of 17.800 usec -> 2015-09-18T14:08:56.465.120.550 (XO: 1.4652618)
   Max dTM/dt [1/us]::9.30e+01 at an offset of 239.826 usec -> 2015-09-18T14:08:56.465.342.575 (XO: 1.4654838)
   Min dTM/dt [1/us]::-9.50e+01 at an offset of 84.200 usec -> 2015-09-18T14:08:56.465.186.950 (XO: 1.4653282)

 - Block number: 2 starting at 2015-09-18T14:09:16.740.258.750 (XO: 21.7404006, t=35.6°C)  Rank: 00003
   TM values: Max =    10  Min =   -18  Ave =    -4.65  Std =     3.29
   Min TM value:-18 at an offset of 95.175 usec -> 2015-09-18T14:09:16.740.353.925 (XO: 21.7404957)
   Max TM value:10 at an offset of 80.800 usec -> 2015-09-18T14:09:16.740.339.550 (XO: 21.7404814)
   Max dTM/dt [1/us]::9.10e+01 at an offset of 195.863 usec -> 2015-09-18T14:09:16.740.454.612 (XO: 21.7405964)
   Min dTM/dt [1/us]::-9.60e+01 at an offset of 194.988 usec -> 2015-09-18T14:09:16.740.453.737 (XO: 21.7405955)

 - Block number: 3 starting at 2015-09-18T14:09:35.307.570.750 (XO: 40.3077127, t=35.6°C)  Rank: 00003
   TM values: Max =     6  Min =   -16  Ave =    -4.64  Std =     2.97
   Min TM value:-16 at an offset of 47.800 usec -> 2015-09-18T14:09:35.307.618.550 (XO: 40.3077605)
   Max TM value:6 at an offset of 19.713 usec -> 2015-09-18T14:09:35.307.590.462 (XO: 40.3077324)
   Max dTM/dt [1/us]::8.60e+01 at an offset of 105.950 usec -> 2015-09-18T14:09:35.307.676.700 (XO: 40.3078187)
   Min dTM/dt [1/us]::-8.90e+01 at an offset of 35.388 usec -> 2015-09-18T14:09:35.307.606.137 (XO: 40.3077481)
