-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20150915_222325_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 10.298898 to 35.649011
-each separately from 10s to 35s and from 298898us to 649011us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.145731 in 1 packets - #/No: 18/18
      11.203515 in 1 packets - #/No: 36/18
MIU packet time from 2015-09-15T22:23:38.298.898.000 to 2015-09-15T22:24:03.649.011.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.145731 in 1 packets - #/No: 18/18
      11.203515 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.2837100 to 37.2836469
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999647 in 1 packets - #/No: 18/18
      10.9999722 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      08 -> 44.6°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20150915_222325_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.5826086 to 12.5828976
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.1456950  -- OK for SL --
   MIU time from 2015-09-15T22:23:38.298.898.000 to 2015-09-15T22:23:38.299.187.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.145731

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.7285926 to 26.7288816
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.2034870  -- OK for SL --
   MIU time from 2015-09-15T22:23:52.444.918.000 to 2015-09-15T22:23:52.445.207.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.203515

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.9323686 to 37.9326576
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-09-15T22:24:03.648.722.000 to 2015-09-15T22:24:03.649.011.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-09-15T22:23:38.298.757.750
   Max SL corrected packet time: 2015-09-15T22:24:03.648.870.750
   Min SL corrected XO rel time: 10.2988994
   Max SL corrected XO rel time: 35.6490123
   TM values from -38 to 29 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-09-15T22:23:38.298.757.750 (XO: 10.2988994, t=34.0°C)  Rank: 00005
   TM values: Max =    21  Min =   -32  Ave =    -4.82  Std =     7.21
   Min TM value:-32 at an offset of 108.225 usec -> 2015-09-15T22:23:38.298.865.975 (XO: 10.2990076)
   Max TM value:21 at an offset of 101.188 usec -> 2015-09-15T22:23:38.298.858.937 (XO: 10.2990006)
   Max dTM/dt [1/us]::3.98e+02 at an offset of 102.113 usec -> 2015-09-15T22:23:38.298.859.862 (XO: 10.2990015)
   Min dTM/dt [1/us]::-3.93e+02 at an offset of 90.700 usec -> 2015-09-15T22:23:38.298.848.450 (XO: 10.2989901)

 - Block number: 2 starting at 2015-09-15T22:23:52.444.777.750 (XO: 24.4449191, t=34.0°C)  Rank: 00007
   TM values: Max =    29  Min =   -38  Ave =    -4.83  Std =     8.51
   Min TM value:-38 at an offset of 108.675 usec -> 2015-09-15T22:23:52.444.886.425 (XO: 24.4450277)
   Max TM value:29 at an offset of 125.275 usec -> 2015-09-15T22:23:52.444.903.025 (XO: 24.4450443)
   Max dTM/dt [1/us]::5.47e+02 at an offset of 125.238 usec -> 2015-09-15T22:23:52.444.902.987 (XO: 24.4450443)
   Min dTM/dt [1/us]::-5.33e+02 at an offset of 124.350 usec -> 2015-09-15T22:23:52.444.902.100 (XO: 24.4450434)

 - Block number: 3 starting at 2015-09-15T22:24:03.648.581.750 (XO: 35.6487233, t=34.0°C)  Rank: 00006
   TM values: Max =    24  Min =   -33  Ave =    -4.83  Std =     7.69
   Min TM value:-33 at an offset of 124.125 usec -> 2015-09-15T22:24:03.648.705.875 (XO: 35.6488474)
   Max TM value:24 at an offset of 124.200 usec -> 2015-09-15T22:24:03.648.705.950 (XO: 35.6488475)
   Max dTM/dt [1/us]::5.08e+02 at an offset of 124.163 usec -> 2015-09-15T22:24:03.648.705.912 (XO: 35.6488475)
   Min dTM/dt [1/us]::-4.23e+02 at an offset of 125.050 usec -> 2015-09-15T22:24:03.648.706.800 (XO: 35.6488484)
